DATE NOTIFICATION INITIATED:
An error was contained on the ETI-LAB Applications disk for programming the BioRad HIV-1/HIV-2 Plus O assay. The error induced specimen and conjugate incubation temperatures for the assay to remain at ambient temperature rather than the required 37°C temperature. Consignees will receive replacement disks.
The information in this listing reflects CBER's best efforts to communicate information that has been reported to FDA. Its accuracy and comprehensiveness cannot be guaranteed.